Download HP A RF IMPEDANCE-ANALYZER OPERATION SM service manual & repair info for electronics experts. The HP A RF Impedance Analyzer measures 14 parameters Parameter measured: IZI, Swept measurement: manual or automatic sweep from start to. Agilent A Test Fixture. Operation and Service Manual. Third Edition. Agilent Part No. February Printed in Japan.
|Published (Last):||11 March 2010|
|PDF File Size:||11.64 Mb|
|ePub File Size:||18.13 Mb|
|Price:||Free* [*Free Regsitration Required]|
Good luck with it I have worked with the A and, as I recall, the commands are nothing like any other instrument I have seen. In the code above, MHz is an arbitrary frequency.
Keysight / Agilent A – In Stock, We Buy Sell Repair, Price Quote
Guess you never know until you try! I have not worked with that instrument, but the Agilent web site has the operating and service manual here. I am able to mnual with the device through the GPIB interface, and am also able to retrieve information. IIRC, the sweep parameters were something like this: Should I be looking at doing it with LabView or something else? Has anyone done anything on interfacing with the A since Hhp4191a C’s last post?
Solved: GPIB Coding for the HPA Impedence Analyzer – Discussion Forums – National Instruments
If not, what would creating a driver for it entail? I think the would probably take the old commands, manaul the ALSO has new commands. Matt – you’re right, the commands for the A are completely different from the – probably because of its age. Would you mind posting the code you used on the A?
However, rather than being returned the data at MHz, I am returned the data at whatever frequency the analyzer is currently set at. In particular, after setting up and initiating the instrument, I am initiating the measurement and reading it as follows:.
Message 6 of 6.
Message 4 of 6. I went through the manual before, but when I saw their GPIB section, the strings were so convoluted and so unlike anything I’ve ever seen before that I assumed it was referring to something different. Message 5 of 6. Message 3 of 6.
Most Active Software Boards: When I gave it a try, though, it worked fine. Accepted by topic author Matt C. Thanks for your help guys.
Message 2 of 6. In particular, after setting up and initiating the instrument, I am initiating the measurement and reading it as follows: It’s 84MB, so I hope you have a fast link.
Message 1 of 6. Sorry, that last line of code there is supposed to read: You really need the manual.
Agilent / HP 4191A RF Impedance Analyzer
Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type. I suspect that this difference might be causing my problems.
The forums emoticons took over for a bit there.